This Broadley-James D635-VP (25mm Sartorius) Dissolved Oxygen Sensor is designed for use with sartorius ports which are slightly longer than the standard length and have a different internal design. As such they need slightly different sensors. This VarioPin version has a right-angled connector and is also available as the D630-VP with straight connector. In both of these sensors the o-ring has been moved closer to the tip in order to create a better seal. It is also available in both formats with standard D9 connectors.
The sensor fits directly into the vessel, eliminating the need for a housing, reducing contamination risks. Few accessories are required. The modular design allows quick replacement of any part, making it easy to maintain and repair the sensor. OxyProbe dissolved oxygen sensors carry a 1-Year Warranty.
D635-VP (25mm Sartorius) Dissolved Oxygen Sensor Specification
- 316L Stainless Steel
- Serialized for traceability
- RA32 finish with electropolish on wetted parts
- External EPDM o-ring
- Secured retainer ring
Broadley-James offer a range of membrane kits to suit all our dissolved oxygen sensors. These include everything required to replace the membrane cartridge and internal o-rings. All kits include:
- QC Certificate
- Material Cert
Use our Membrane Replacement instructions for detailed information about replacing the membrane in the model D635-VP (25mm Sartorius) Dissolved Oxygen Sensor
The following membrane kits are suitable for use with the D635-VP (25mm Sartorius) Dissolved Oxygen Sensor:
The D635-VP (25mm Sartorius) Dissolved Oxygen Sensor uses a VP (VarioPin) connector. This multifunctional electrode is capable of both receiving and processing a number of signals at the same time. We recommend the following cables for use with the D635-VP Dissolved Oxygen sensor:
The internal seals of an Oxyprobe are all mechanical seals using o-rings. There are no adhesives to leak or fail in process, making this a more rugged and reliable sensor. Replacing the cathode is also much easier and faster since there is no sealant to clean from the threads. No sealant also makes the job of replacing the cathode much faster and cleaner.