This 19mm Dissolved Oxygen Sensor is for use in Novaseptic installations and in 1.5″ sanitary tees. It is important to note, these sensors are not intended for use with the standard 25 mm sanitary flange ports. However, they can be used with the NovAseptic ports.
The interior design and replacement parts of this sensor are the same as other 25 mm OxyProbes. The modular design allows quick replacement of any part, making it easy to maintain and repair the sensor. OxyProbe dissolved oxygen sensors carry a 1-year warranty.
This sensor is also available with right-angled connector as D215.
- 316L Stainless Steel
- Serialized for traceability
- RA32 finish with electropolish on wetted parts
- Fits in both sanitary flange ports and NovAseptic ports.
- Can also be used in-line in a sanitary pipe tee.
It is important to maintain the sensors by inspecting and calibrating them regularly. The membrane cartridge should be checked and refilled prior to every use.
All OxyProbe dissolved oxygen sensors are available in both straight and right-angled connector versions. This low profile design provides:
- Protection against cable fatigue
- Improved grip for sensor extraction
- Reduced random sensor mobility on flat surfaces (won’t roll off the table)
D210 (19mm) Dissolved Oxygen Sensor Specification
- Measurement principal: Amperometric – Polarographic
- Output signal in air: Approx. 67 nA at 25°C
- Temperature range: Operation: 0 to 80°C
Sterilisation: Max. 135°C
- Stability: Better than 2% of readout per week at constant pressure and temp. (30°C)
- Maximum Pressure: 4 bar (atm.), 58 psig
The Models D210 Series Dissolved Oxygen OxyProbe sensors are intended for use in a 1 1/2” sanitary pipe tee. The sensor is secured into the tee with a standard flange clamp sealed with a flange gasket. The tip of the sensor is positioned in the middle of the process flow when it is placed in a standard size tee. These sensors can also be used in NovAseptic ports.
The Model D210 or D215 sensor installed in a NA-connect® port offers potential benefits in biotech applications. The unique design of the port offers increased drainage around the port and most effective Clean-In-Place (CIP). This port, which is almost flush to the vessel wall, allows the use of a 19 mm sensor front end, since there is no need for a forward o-ring to seal the sensor into the vessel. Minimizing or eliminating the cracks or crevices improves clean-ability.
Sometimes found on vessels in biotech facilities, this style of port is flush to the vessel wall to minimize crevices. The unique design of the port offers increased drainage around the port and more effective Clean-In-Place (CIP). The NA-connect® port from NovAseptic is compatible with a special sanitary housing, which is secured to the port by a clamp. The port is installed at a 15° angle to allow for proper electrode performance. See the cutaway drawing to the right for typical installation dimensions.
The internal seals of an Oxyprobe are all mechanical seals using o-rings. There are no adhesives to leak or fail in process, making this a more rugged and reliable sensor. Replacing the cathode is also much easier and faster since there is no sealant to clean from the threads. No sealant also makes the job of replacing the cathode much faster and cleaner.